Am 10.09.2010 21:15, schrieb János, Tóth F.:
By the way, I think you didn't really fixed the "Sample Read failed" bug or there is no solution. I used to receive this error during display calibration and I never noticed it during the characterization. The calibration was stable now but I received this familiar message during the characterization (once with both of my displays). I think it can be a problem with the instrument black self-calibration because I used DisplcalGUI and I let it to bypass the instrument self-calibration between the calibration and characterization. (And I chose the High quality calibration, so it took a lot of time...)
This was also reported to me, so the default in the upcoming next version of dispcalGUI will be to not skip the instrument calibration for the characterization readings - as I understand it, this solves the issue in that case (existing users still have to turn off the respective menu option in dispcalGUI though, just like now). Btw, I'm preparing a new snapshot (with controls for the new features in Argyll CMS 1.3.0), but ETA is still a few hours maybe.
Regards -- Florian Höch