[argyllcms] Re: Trouble getting Argyll 1.8.3 to recognize ColorMunki Design (Windows 10)
- From: Graeme Gill <graeme@xxxxxxxxxxxxx>
- To: argyllcms@xxxxxxxxxxxxx
- Date: Tue, 12 Jul 2016 09:51:26 +1000
Chris Lilley wrote:
Argyll seems not to recognize the device, setting the rotary dial to the
position is no different to any other position.
sounds like your instrument isn't working properly. Some Munki's
have been reported to play up in regard to the rotary dial
position recognition after some use, and one in my possession had
the flexible PCB that connects the USB to the main electronics fail
(something that I managed to fix by replacing it).
Trying again with the default windows driver installed (Shows up in device
X-Rite devices > Colormunki)
PS C:\Users\Chris> dispcal -r
Argyll 'V1.8.3' Build 'MSWin 64 bit' System 'Windows V10.0 SP 0'
dispcal: Error - No instrument at port 1
Well of course - ArgyllCMS can't use X-Rite's system driver, it
expects the libusb-win32 driver. You need to run X-Rite's whole
software stack if you want that to work (System driver, Instrument
access service, Application).
Other related posts: