I just wanted to say a big thanks to Graeme for all his help in getting my
non-standard instrument working. Needless to say a donation has been made.
I will look to keep the device as the seller has offered a partial refund.
Whilst it's not a fully featured i1 Pro 2, I'm hoping that I can still get good
results with it. If nothing else, I can sell my Colormunki Photo and be left
with a slightly more capable device for not much more outlay.
Thanks all :-)
Sent from Outlook<http://aka.ms/weboutlook>
From: argyllcms-bounce@xxxxxxxxxxxxx <argyllcms-bounce@xxxxxxxxxxxxx> on behalf
of Graeme Gill <graeme@xxxxxxxxxxxxx>
Sent: 07 December 2017 00:46
Subject: [argyllcms] Re: i1 Pro 2 Problems
Summarizing the resolution of this topic:
* The i1Pro2 in question seems to be a stripped down OEM instrument,
missing many of the i1Pro2 features. My guess is that it is an OEM
i1Pro Rev D replacement. The most noticeable limitations
are lack of Ambient measurement, and the presence of a UV filter.
Less visible differences are lack of a Wavelength Calibration LED,
lack of Zebra Ruler stripe support, and lack of UV illumination LED
(which would be useless anyway, given the UV filter.)
Anybody buying a second hand i1Pro2 might want to be aware
that some of these are floating around.
* ArgyllCMS V2.0.1 when released, will be able to work with the instrument.