Nikolay Pokhilchenko wrote:
I've noticed many times that chartread with i1Pro may misread patches when there are non-contrast or too narrow spacers between patches. It likely may took a place in the case when there was no right preliminary profile for *.ti2 data and printtarg places patches unoptimal way. Another reason to misread may be a blank paper strip between patches and sheet edge if the edge is crossed by instrument while reading. In such case the strip cut off straight on patch column edge may help.
When you say "misread", do you mean misread without noticing the problem, or do you mean there is a reported misread ? I haven't seen any examples of non-reported misreads, with the exception of the direction detection problem I mentioned before. I'd be interested in hearing about any other reproducible examples of non-reported misread problems. Graeme Gill.