Vallo Kallaste wrote:
Vallo Kallaste wrote:Dispcal -v -yc -R is consistently giving white level reading as 2xxx cd/m², which seems to be incorrect or am I misunderstanding something? Using Spyder3 and measuring Sony G420 CRT. Dispcal version is 1.1.0RC2 but the win32-usb is what came with an early 1.1.0 beta because I already have it. Datacolor software reading is 144 cd/m².
Hi, I think this is a bug in my driver, although it is triggered by a change in the Spyder 3 calibration data. Try the spotread.exe in <http://www.argyllcms.com/spotread.zip> and see if it is better. Please mail me a "-D5" output too. Thanks, Graeme Gill.