Vallo Kallaste wrote:
Dispcal -v -yc -R is consistently giving white level reading as 2xxx cd/m², which seems to be incorrect or am I misunderstanding something? Using Spyder3 and measuring Sony G420 CRT. Dispcal version is 1.1.0RC2 but the win32-usb is what came with an early 1.1.0 beta because I already have it. Datacolor software reading is 144 cd/m².
Are you able to reproduce the problem with the 1.1.0 RC2 binaries ? (See <http://www.freelists.org/post/argyllcms/ArgyllCMS-V110-RC2-is-now-available>) ? If so, please send me the "spotread -D5 -yc" of a white reading. Graeme Gill.