Magnus Wallden wrote:
The Spectrolino "mode" dont work, for this it need a external power supply, or a 8p minidin extension for power from spectroscan.
Right. According to the manual, Pin 7 (of the 8 pin) is +10 to +24 Volts at 4 Watts, and Pin 8 is Power Ground.
that whit about 5-6 mm dia outline and a hole for a pin, but no polarity and no voltage.
The normal power supply is 15V DC @ 0.8A, Center is +ve, but from the above, 12 Volts will probably work too. Graeme Gill.