Beisch Clemens wrote:
I tried using the iOne Rev.A with dispcal on the following setups: 1. Powerbook G4 with OS X 10.4.8 2. Powermac G5 with OS X 10.4.9
Thanks for your feed back. I think this is a process scheduling/latency issue, which causes the measurement reading to fail. I've figured out a way of avoiding this, so I'm expecting the next beta release will cure this problem. Graeme Gill.