Leonard Evens wrote:
leaving the previous settings for brightness, constrast adn R, G, and B. I had to do it three times to get a profile. The first time it hung on patch 13 of 64 under iteration 3, and the second time it hung on patch 26, eventually timing out and reporting a communication error with the device.
Hi, depending on how repeatable the problem is, the thing to do is to grab a debug trace and send it to me. ie. dispcal -v2 -D5 [other options & parameters] 2>&1 > log
I have used this Eye One Pro repeatedly on another (32 bit Dell) computer without incident. Has anyone any suggestions about what the problem may related to: e.g, the 64 bit system, the particular USB port, or what?
Anything is possible. Did you compile Argyll from source or download the binaries from argyllcms.com ? Unfortunately downstream distributions insist on meddling with the code, and it often breaks things. One thing they tend to break is the USB drivers, since they substitute the system libusb and don't test their changes with all the instruments. It's possible the kernel level USB drivers are subtly different, or it's possible that there is a bug (instrument/USB driver interactions that aren't handled properly). cheers, Graeme Gill.