Lars Tore Gustavsen wrote:
This is direct from the Scenarios.html
Optimising layout for strip reader: 100% After optimisation, worst delta E = 80.126192 [ltg@hjemme r2400]$ Can this be correct? When I try to other instruments with the -i option I don't get the "worst case delta E" line at all, but more like this:
Worst case density contrast = 0.701037 Optimising layout for strip reader: 100% After optimisation, density contrast = 0.814464
This is just verbose output. Some strip reading instruments detect patches according to density contracts, others by delta E. Printtarg attempts to optimize the patch to patch/spacer contrast according to the instrument type. The DTP41 looks for density contrast, the i1pro looks for delta E between patches/spacers, so printtarg behaves appropriately. Graeme Gill.