[argyllcms] Re: Help me format a printer calibration .ti3 file?
- From: Graeme Gill <graeme@xxxxxxxxxxxxx>
- To: argyllcms@xxxxxxxxxxxxx
- Date: Wed, 6 Dec 2017 09:40:51 +1100
TONY ARNERICH wrote:
results on a ColorChecker didn't change by more than 1 dE from the previous
housing, but the painting will happen anyway.
Another comment :- having a single light source direction will be fine for
colored surfaces (this is what the ColorMunki spectro does), but something
like the i1Pro and other more expensive instruments use a ring mirror to
illuminate at 45 degrees from all directions. This eliminates dependence on
the orientation of the instrument when measuring textured surfaces.
Other related posts: