Kompatscher Klaus wrote:
using the attached data and creating a profile with calprof I get a white point of L86 a-8 b0which does not correspond to the real media white.
Hi, How are you determining the "real" media white ? The single white patch in the data set has a value of Lab 85.059164 -8.124163 -0.490875 using D50 illuminant and the 1931 2 degree observer, while the smoothed profile white point is 86.311126 -8.360063 -0.733073, a difference of 1.296812 delta E. This seems perfectly normal. Graeme Gill.