Nikolay Pokhilchenko wrote:
Is the i1Display LT temperature stabilized?
Not explicitly, but I think the LtoF chip they use has a relatively low temperature coefficient (ie. it may do some internal compensation).
I have both i1 Display LT and i1Pro rev.D non-UVCut. Can I make correction matrix for i1Display and calibrate/profile the display by i1 Display with CCMX instead calibrating/profiling by i1 Pro? Can be this way more accurate for wide-gamut displays? For sRGB gamut displays? What do You think?
Yes, if you have a spectrometer as a reference, you can certainly create a ccmx for the colorimeter. Graeme Gill.