Fabrizio Levati wrote:
Is there a way to investigate the problem further?
Not without some inside knowledge of the instrument, or a lot of messing about.
I'm asking because the instrument is working properly with at least two softwares... Sorry for the dumb question, could it be that, being re-branded by quato, it is someway non-standard?
I wouldn't have thought so. It's returning the error in response to the reset command, which is one of the most basic commands. I guess it might still operate if the error is ignored, it just won't give good results if there is no temperature drift calibration. So even if other software manages to work, I wouldn't trust the readings from it. Graeme Gill.