Simon Kirby wrote:
dispwin is unhappy with My DPT94 when I try 1.3.0 binaries, source, or a build of 1.3.1 beta. Debian sid 1.1.1 package seems to work, as does my ancient 0.70 beta 8 build. 1.1.1: [sroot@oof:~]# dispcal -K -H -m -g2.2 -yl -q h -v -o acd XRandR 1.2 is faulty - falling back to older extensions Setting up the instrument Instrument Type: DTP94 Serial Number: 109477 Boot version: D929 Software version: DB06 Place cap on the instrument, or place in on a dark surface, or place on the white calibration reference, and then hit any key to continue, or hit Esc or Q to abort: 1.3.0: [sroot@oof:/usr/src/argyll/Argyll_V1.3.0/bin]# ./dispcal -K -H -m -g2.2 -yl -q h -v -o acd XRandR 1.2 is faulty - falling back to older extensions Setting up the instrument dispcal: Error - docalibration failed with return value -1
Sorry, I'm unable to reproduce this. First try simplifying your options. (The DTP94 doesn't support the -H option for instance. It's also probably not a good idea to put -o as the last option either, since it has an optional file argument.) Try: ./dispcal -v -yl acd and see if it talks to the instrument. Then try: ./dispcal -v -yl -K acd and see if you get an error. If you do get a problem, then you can help diagnose it by using -D: ./dispcal -v -D5 -K acd [BTW: you need to subscribe to this list and post from the email address you subscribed from, for your posts to make it to the list.] Graeme Gill.