adam k wrote:
Am I to understand that CMunki is better than dtp94? Should I replace dtp94 then to calibrate my u2410 and to make printer profiles?
They are different types of instruments. The dtp94 is a colorimeter, while the ColorMunki (Design or Photo) is a spectrometer. [Note that the ColorMunki Create is a version of the i1display 2 colorimeter.] A colorimeter depends on its physical filters being a match to the observer curves, and typically in low cost (non-laboratory) instruments we're talking about here, the match isn't very good, hence all the problems with wide gamut monitors. A spectrometer analyzes the spectrum of the device, so the observer curves are implemented in software, and are a lot more accurate, hence a spectrometer can deal with wide gamut monitors well. The ColorMunki is also much more flexible device. Graeme Gill.