Graeme Gill skrev 2012-01-30 23:20:
Kristian Jörg wrote:Hmm, one of their selling points (CP) is that the meter has very stable readings, even in very low light. Maybe they are reading several measurements and getting an average?Longer readings have less variation due to refresh, but it is only a linear improvement with integration time. If the reading synchronises, you don't have this trade-off. Beware that claims of of improved stability may be assuming non-refresh conditions (LCD).
Ok. Is plasma regarded as a refresh display or not?
Do you provide standard calibration files with the i1d3 driver for Argyll?I'm not sure what you mean by this. Only one calibration comes with Argyll, and that is for a CRT like display. You can use the i1d3ccss to access the display calibrations that come with your instrument. <http://www.argyllcms.com/doc/i1d3ccss.html>
Ok, excellent! /Kristian