andrzej duda wrote:
Nothing wrong with the numbers.But after the profile was installed I've validated it with basICColor display 4.1.17and got only 71 cd/m2 instead of 100. This is rather huge difference. Any idea why?
Which instrument ? Which version of Argyll ? There is a bug in V1.0.4 for the i1pro in which the absolute brightness has an error of 1.37. Note that 1.37 * 71 = 97.3. Graeme Gill.