When do you expect a new build (RC3?) will be posted with this fix in it? How close are you getting to the release of 1.1.0 final?
Well both depend on how many problems/minor features people keep asking for :-) I was hoping to get at RC3 out in the next couple of days.
Is there a setting for integration time in Argyll?
Not without recompiling the i1pro driver. For the display mode it chooses an integration time based on the white brightness, since too long an integration time will cause sensor overload. For a screen of less than 110 cd/m^2 it uses 1 reading of 2 seconds. For less than 275 cd/m^2 2 readings of 0.8 seconds, for less than 700 cd/m^2 3 readings of 0.3 seconds.
If you create a test build with that adaptive reading mode enabled, feel free to email it to me an I'll test it out.
I think RC3 will be available soon. Graeme Gill.