Kompatscher Klaus wrote:
If I use the "Auflicht" data to creat a profile with PM5 I get L87 a 0 b-2.4 as white point. Something went wrong but I have no clue where or what.
You have a corrupted .txt file. At line 915 the value for the 610nm is "$0.3752". This throws off the guessing of the field type, and all the 610nm values are set to zero as a result. I'll fix logo2cgats so that it checks the spectral type and errors out rather than ignoring this. Graeme Gill.