Am 06.12.2006 um 23:09 schrieb Gerhard Fuernkranz:
Graeme Gill wrote:Burckhart Seifert wrote:After analyzing [...]
Most of the profile fit errors seem to be near cyan == 100%. Something funny seems to be going on in that channel at least, as it approaches 100%. Is there a calibration system on the RIP ? How are the calibration curves targeted ? The chart is very regularly arranged,Is this possibly an ECI chart?
Yes, it is the ECI2002. I have also made it available. http://www.colortaxi.de/misc/argyll/The linearization is made within the RIP. I detected the maximum density for the cyan channel at 97%.
Regards, Gerhardwhich is often not very efficient (that's why Argyll generate charts are different).
Thanks for looking at it. Burckhart SeifertPS: What I also seem to have to do is to replace the--with the newest builds--only updated RIP software by a complete new installation of the actual version.