Mickael Profeta wrote:
Black = XYZ 999.99 0.00 999.99 Red = XYZ 31.09 14.61 0.22 Green = XYZ 24.39 47.13 6.82 Blue = XYZ 12.22 3.31 59.62 White = XYZ 69.46 70.51 75.29
I attach the log file
Hmm. This indicates that the 999.99 is coming from the instrument, so something is a bit strange about it.
The Y value of 0.0 is still highly suspicious, so the 999.99 could be a rollover value. Try calibrating the instrument again. Make sure it is completely in the dark for the black level calibration (ie., inside it's plastic "holder", wrapped in something opaque).
If this doesn't fix it, perhaps it needs resetting to factory default. I'm not sure how best to go about that, apart from using a terminal and manually typing in "5CRI".