Rishi Sanyal wrote:
Doesn't your adaptive mode help with the spectrophotometer (or 'spectroradiometer')? Or can the adaptive mode only help to a certain extend? I understand that colorimeters with matched filters can be
Adaptive helps the accuracy, but it doesn't help stability. The main drawback of the current i1Pro and Munki spectrometers is that they aren't temperature compensated when in emissive mode, while many of the colorimeters are temperature stabilised. My dispcal/dispread -Ib option is an attempt to work around this issue with the current spectrometers. (I'd expect the really expensive spectrometers to be temperature compensated though.) Graeme Gill.